A weather station is a facility, either on land or sea, with instruments and equipment for measuring atmospheric conditions to provide information for weather forecasts and to study the weather and climate. The measurements taken include temperature, atmospheric pressure, humidity, wind speed, wind direction, and precipitation amounts. Wind measurements are taken with as few other obstructions as possible, while temperature and humidity measurements are kept free from direct solar radiation, or insolation. Manual observations are taken at least once daily, while automated measurements are taken at least once an hour. Weather conditions out at sea are taken by ships and buoys, which measure slightly different meteorological quantities such as sea surface temperature (SST), wave height, and wave period. Drifting weather buoys outnumber their moored versions by a significant amount.
Typical weather stations have the following instruments:
In addition, at certain automated airport weather stations, additional instruments may be employed, including:
Except for those instruments requiring direct exposure to the elements (anemometer, rain gauge), the instruments should be sheltered in a vented box, usually a Stevenson screen, to keep direct sunlight off the thermometer and wind off the hygrometer. The instrumentation may be specialized to allow for periodic recording otherwise significant manual labour is required for record keeping. Automatic transmission of data, in a format such as METAR, is also desirable as many weather station's data is required for weather forecasting.
A personal weather station is a set of weather measuring instruments operated by a private individual, club, association, or business (where obtaining and distributing weather data is not a part of the entity's business operation). Personal weather stations have become more advanced and can include many different sensors to measure weather conditions. These sensors can vary between models but most measure wind speed, wind direction, outdoor and indoor temperatures, outdoor and indoor humidity, barometric pressure, rainfall, and UV or solar radiation. Other available sensors can measure soil moisture, soil temperature, and leaf wetness. The quality, number of instruments, and placement of personal weather stations can vary widely, making the determination of which stations collect accurate, meaningful, and comparable data difficult. There are a comprehensive number of retail weather stations available.
Personal weather stations typically involve a digital console that provides readouts of the data being collected. These consoles may interface to a personal computer where data can be displayed, stored, and uploaded to websites or data ingestion/distribution systems. Open-source weather stations are available that are designed to be fully customizable by users.
Personal weather stations may be operated solely for the enjoyment and education of the owner, while some owners share their results with others. They do this by manually compiling data and distributing it, distributing data over the Internet, or sharing data via amateur radio. The Citizen Weather Observer Program (CWOP) is a service which facilitates the sharing of information from personal weather stations. This data is submitted through use of software, a personal computer, and internet connection (or amateur radio) and are utilized by groups such as the National Weather Service (NWS) when generating forecast models. Each weather station submitting data to CWOP will also have an individual Web page that depicts the data submitted by that station. The Weather Underground Internet site is another popular destination for the submittal and sharing of data with others around the world. As with CWOP, each station submitting data to Weather Underground has a unique Web page displaying their submitted data. The UK Met Office's Weather Observations Website (WOW) also allows such data to be shared and displayed.
To correctly gather accurate data from a personal weather station the user must set it up correctly with some rules in mind. The user needs to place the station at least 5 meters away from the nearest object, if that object is over 5 meters tall then the distance must be 10 meters minimum. Ensure the station is not in an alleyway as the wind tunnel effect can lead to inaccurate readings on the anemometer. Following these rules will lead to a higher accuracy in readings.
Home weather stations include hygrometers, thermometers, barographs, and barometers. Commonly wall mounted and made by manufacturers such as Airguide, Taylor, Springfield, Sputnik and Stormoguide.
A weather ship was a ship stationed in the ocean as a platform for surface and upper air meteorological measurements for use in weather forecasting. It was also meant to aid in search and rescue operations and to support transatlantic flights. The establishment of weather ships proved to be so useful during World War II that the International Civil Aviation Organization (ICAO) established a global network of 13 weather ships in 1948. Of the 12 left in operation in 1996, nine were located in the northern Atlantic ocean while three were located in the northern Pacific ocean. The agreement of the weather ships ended in 1990. Weather ship observations proved to be helpful in wind and wave studies, as they did not avoid weather systems like merchant ships tended to and were considered a valuable resource. The last weather ship was MS Polarfront, known as weather station M ("jilindras") at 66°N, 02°E, run by the Norwegian Meteorological Institute. MS Polarfront was removed from service January 1, 2010. Since the 1960s this role has been largely superseded by satellites, long range aircraft and weather buoys. Weather observations from ships continue from thousands of voluntary merchant vessels in routine commercial operation; the Old Weather crowdsourcing project transcribes naval logs from before the era of dedicated ships.
Weather buoys are instruments which collect weather and oceanography data within the world's oceans and lakes. Moored buoys have been in use since 1951, while drifting buoys have been used since the late 1970s. Moored buoys are connected with the seabed using either chains, nylon, or buoyant polypropylene. With the decline of the weather ship, they have taken a more primary role in measuring conditions over the open seas since the 1970s. During the 1980s and 1990s, a network of buoys in the central and eastern tropical Pacific ocean helped study the El Niño-Southern Oscillation. Moored weather buoys range from 1.5-12 metres (5-40 ft) in diameter, while drifting buoys are smaller, with diameters of 30-40 centimetres (12-16 in). Drifting buoys are the dominant form of weather buoy in sheer number, with 1250 located worldwide. Wind data from buoys has smaller error than that from ships. There are differences in the values of sea surface temperature measurements between the two platforms as well, relating to the depth of the measurement and whether or not the water is heated by the ship which measures the quantity.
Synoptic weather stations are instruments which collect meteorological information at synoptic time 00h00, 06h00, 12h00, 18h00 (UTC) and at intermediate synoptic hours 03h00, 09h00, 15h00, 21h00 (UTC).
The common instruments of measure are anemometer, wind vane, pressure sensor, thermometer, hygrometer, and rain gauge.
The weather measures are formatted in special format and transmit to WMO to help the weather forecast model.
A variety of land-based weather station networks have been set up globally. Some of these are basic to analyzing weather fronts and pressure systems, such as the synoptic observation network, while others are more regional in nature, known as mesonets.
The firmware is under active development, and users are encouraged to contribute to it by forking the repository on Github. An initiative to further develop the hardware and make it commercially available is ongoing as spin-off project OpenObservatory.