In optics, absorbance or decadic absorbance is the common logarithm of the ratio of incident to transmitted radiant power through a material, and spectral absorbance or spectral decadic absorbance is the common logarithm of the ratio of incident to transmitted spectral radiant power through a material. Absorbance is dimensionless, and in particular is not a length, though it is a monotonically increasing function of path length, and approaches zero as the path length approaches zero. The use of the term "optical density" for absorbance is discouraged. In physics, a closely related quantity called "optical depth" is used instead of absorbance: the natural logarithm of the ratio of incident to transmitted radiant power through a material. The optical depth equals the absorbance times ln(10).
The term absorption refers to the physical process of absorbing light, while absorbance does not always measure absorption: it measures attenuation (of transmitted radiant power). Attenuation can be caused by absorption, but also reflection, scattering, and other physical processes.
Absorbance of a material, denoted A, is given by
Absorbance is a dimensionless quantity. Nevertheless, the absorbance unit or AU is commonly used in ultraviolet-visible spectroscopy and its high-performance liquid chromatography applications, often in derived units such as the milli-absorbance unit (mAU) or milli-absorbance unit-minutes (mAU×min), a unit of absorbance integrated over time.
Absorbance is related to optical depth by
where ? is the optical depth.
Spectral absorbance in frequency and spectral absorbance in wavelength of a material, denoted A? and A? respectively, are given by
Spectral absorbance is related to spectral optical depth by
Although absorbance is properly unitless, it is sometimes reported in "arbitrary units", or AU. Many people, including scientific researchers, wrongly state the results from absorbance measurement experiments in terms of these made-up units.
Absorbance is a number that measures the attenuation of the transmitted radiant power in a material. Attenuation can be caused by the physical process of "absorption", but also reflection, scattering, and other physical processes. Absorbance of a material is approximately equal to its attenuance[clarification needed] when both the absorbance is much less than 1 and the emittance of that material (not to be confused with radiant exitance or emissivity) is much less than the absorbance. Indeed,
that is equivalent to
and according to Beer-Lambert law, , so
Absorbance of a material is also related to its decadic attenuation coefficient by
If a(z) is uniform along the path, the attenuation is said to be a linear attenuation, and the relation becomes
If c(z) is uniform along the path, the relation becomes
The use of the term "molar absorptivity" for molar attenuation coefficient is discouraged.
The amount of light transmitted through a material diminishes exponentially as it travels through the material, according to the Beer-Lambert law (A=(?)(l)). Since the absorbance of a sample is measured as a logarithm, it is directly proportional to the thickness of the sample and to the concentration of the absorbing material in the sample. Some other measures related to absorption, such as transmittance, are measured as a simple ratio so they vary exponentially with the thickness and concentration of the material.
Any real measuring instrument has a limited range over which it can accurately measure absorbance. An instrument must be calibrated and checked against known standards if the readings are to be trusted. Many instruments will become non-linear (fail to follow the Beer-Lambert law) starting at approximately 2 AU (~1% transmission). It is also difficult to accurately measure very small absorbance values (below 10-4) with commercially available instruments for chemical analysis. In such cases, laser-based absorption techniques can be used, since they have demonstrated detection limits that supersede those obtained by conventional non-laser-based instruments by many orders of magnitude (detections have been demonstrated all the way down to 5 × 10-13). The theoretical best accuracy for most commercially available non-laser-based instruments is attained in the range near 1 AU. The path length or concentration should then, when possible, be adjusted to achieve readings near this range.
Typically, absorbance of a dissolved substance is measured using absorption spectroscopy. This involves shining a light through a solution and recording how much light and what wavelengths were transmitted onto a detector. Using this information, the wavelengths that were absorbed can be determined. First, measurements on a "blank" are taken using just the solvent for reference purposes. This is so that the absorbance of the solvent is known, and then any change in absorbance when measuring the whole solution is made by just the solute of interest. Then measurements of the solution are taken. The transmitted spectral radiant flux that makes it through the solution sample is measured and compared to the incident spectral radiant flux. As stated above, the spectral absorbance at a given wavelength is
The absorbance spectrum is plotted on a graph of absorbance vs. wavelength.
A UV-Vis spectrophotometer will do all this automatically. To use this machine, solutions are placed in a small cuvette and inserted into the holder. The machine is controlled through a computer and, once it has been "blanked", automatically displays the absorbance plotted against wavelength. Getting the absorbance spectrum of a solution is useful for determining the concentration of that solution using the Beer-Lambert law and is used in HPLC.
For example, if the filter has 0.1% transmittance (0.001 transmittance, which is 3 absorbance units), its shade number would be 8.